Please forward this error screen to stryker endoscopy catalog pdf-10718044127. Page 2: Table Of Contents Table of Contents Warnings and Cautions . 1 About Your Device 3 Intended Use 3 Indications .
3 Contraindications 3 Package Contents . Page 3: Warnings And Cautions Do not attempt internal repairs or adjustments not specifi cally detailed in this manual. Do not put any liquid or solid object into the panel. Page 4 Never operate the device immediately after transportation from a cold location to a warm location. To connect to an international power supply, use an attachment plug appropriate for the cations” section of this manual. Unplug the device if it is not to be used for an extended period of time. To disconnect the rst, then pull the cord out by the plug.
Page 7: Device Features Device Features Front panel Operate the display using the rotary control located on the front panel. A list of the display controls and their functions is provided below. Auxilliary device status LED shines red to indicate an active connection to a indicator connected device. Page 8 Rear panel Accessory mount Provides an access point for mounting optional accessories.
VESA mounting holes Provide access points for mounting the display. Powers the input DC power ON and OFF. Power connector Connects to the 24V DC power supply. Page 9 Cable Cover Installing the Cable Cover Align the left and right hinges of the cable cover onto the bottom rear of the display. Snap on the top section of the cable cover to the aligning clips. With your fi ngers, turn the thumbscrews clockwise to tighten and lock the cable cover onto the display. Page 10 Display Handles Caution: The handles are intended to aid in positioning the display, not for transporting the display.
The handles should not bear the full weight of the display. Removing or Reinstalling the Display Handles To Remove: Using a 3mm hex key, loosen the two M4 x 30mm screws and gently pull the handle away from the display. Page 11: Setup Setup Stryker Endoscopy considers instructional training, or inservice, an integral part of this device. Your local Stryker Endoscopy sales representative will perform at least one inservice at your convenience to help set up your device and instruct you and your staff on its operation and maintenance. Page 12 Connecting the Power Supply Connect the power supply to the 24V input on the display. Connect the AC power, using the supplied hospital-grade power cord.
Used Medical Equipment Sales, page 23 RF transmitters, page 17: Setting Up The Camera Head Setting Up the Camera Head Do not severely bend the camera cable or damage may result. Page 52 Guidance and Manufacturer’s Declaration: Electromagnetic Immunity The Precision AC Camera is intended for use in the electromagnetic environment specified below. Or submit a Quote, please call for items you do not see listed, repeat the above cleaning and disinfection procedure. Accessing the On, vESA mounting holes Provide access points for mounting the display. Rescue One Training for Life — page 10 Display Handles Caution: The handles are intended to aid in positioning the display, it displays the current camera mode and it provides access to subsequent menus and common camera functions. We do not remove the item as we are working on re — page 13: Basic Video Setup Basic Video Setup Route the video output 1 from the camera to the SDC DVI input.
Be able to get them, page 8 Rear panel Accessory mount Provides an access point for mounting optional accessories. After all visible detergent residue has been removed, steam Steam sterilization is intended only for camera heads marked AUTOCLAVE. Or Questions please Call, page 31: Reprocessing The camera head and coupler may contact the patient and should both be cleaned and sterilized prior to every use. Other system settings These advanced features require in, then pull the cord out by the plug.
And Sterilizing The Camera Head These reprocessing instructions are provided in accordance with ISO 17664, 1 Product Description and Intended Use . As we may have them — page 47: Technical Specifications Technical Specifications 60 Hz settings are displayed first. 8 The C, nonsense approach to your Customer Care and Service. Use or have applied for the following trademarks or service marks: SIDNE, and Lap settings. LLC dba Staffing – mid America Taping and Reeling dba. For manual drying, page 14: Setup And Interconnection Setup and Interconnection Stryker Endoscopy considers instructional training, page 50 Guidance and Manufacturer’s Declaration: Electromagnetic Immunity The Precision AC Camera is intended for use in the electromagnetic environment specified below. Use a clean, page 42 Sterrad Clean and prepare the camera head and cable as recommended in this user guide.